Sigma Spectrum Infusion System v's6.x (model 35700BAX) and Baxter Spectrum Infusion System Version(s) 8.x (model 35700BAX2) at the application layer uses an unauthenticated clear-text communication channel to send and receive system status and operational data. This could allow an attacker that has circumvented network security measures to view sensitive non-private data or to perform a man-in-the-middle attack.
The Baxter Spectrum WBM (v17, v20D29, v20D30, v20D31, and v22D24) telnet Command-Line Interface, grants access to sensitive data stored on the WBM that permits temporary configuration changes to network settings of the WBM, and allows the WBM to be rebooted. Temporary configuration changes to network settings are removed upon reboot.
Baxter Sigma Spectrum Infusion Pumps Sigma Spectrum Infusion System v's6.x model 35700BAX & Baxter Spectrum Infusion System v's8.x model 35700BAX2 contain hardcoded passwords when physically entered on the keypad provide access to biomedical menus including device settings, view calibration values, network configuration of Sigma Spectrum WBM if installed.